This Philips, Respironics Actiwatch, Spectrum Device, Respironics Actigraphy System is used and in fair cosmetic condition. The unit powers on and appears to function properly. The display screen is in fair condition due to several scratches (see photos). All buttons are clean and in good condition. All connection points are clean and in good condition. We do not have the power supply for this unit and are unable to test it further. However, this unit comes with a 30 say satisfaction guarantee. Includes everything in the pictures and nothing else.
Satisfaction Guaranteed or Your Money Back!
Will ship worldwide!
Payment required within 3 days.
DISCLAIMER: Regardless of the origin of the equipment, documentation provided or identification appearing upon the equipment, the equipment described and offered here is in no way certified for, recommended for, or offered for any specific use. The purchaser agrees that the seller shall not be held responsible or liable for any injuries or damages, whether incidental or consequential, associated in any way with the equipment. The purchaser, by bidding on this equipment, indicates their acknowledgment of, and agreement to the terms of this disclaimer.